We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
Page 94 The result will be more accurate if both tests are done at the same blood flow rate. Replacing Freseniue Diasafe Plus Filter 1. The entire Rinse program must be completed before the operator can initiate dialysis.
Page Machine alarms may not occur in every blood loss situation 5.
Fresenius Medical Care 2008K Manuals
Monitoring The Treatment Other Options There are other options on these screens that may not be specifically described here. Chapter 1—Overview Modules The modules accompanying the K are located just below the control panel.
If the external transducer protector and the internal transducer should be become contaminated with blood, the transducer protectors must be replaced, and the transducer and associated parts must be disinfected or replaced.
To shift the window downward, press the Down arrow key. Section 28 — Heat Exchanger Leaking Mwnual Concentrate And Bicarb Pumps Alarm in case of water shortage. Monitor Calibration Procedures Disinfection and Maintenance Disinfection and Maintenance The fluid path of the K can be disinfected chemically or with heat.
Iv Pole And Dialyzer Holder Start key a second time and the pump segment will be rresenius from the pump head.
FRESENIUS MEDICAL CARE K OPERATOR’S MANUAL Pdf Download.
The UF treatment parameters are also saved during a power failure. Following disinfection, they should be rinsed thoroughly with treated water. Chapter 1—Overview The Control Panel The control panel see Figure 3 is located at the top, front of the K machine and contains key pads that allow the user to control the operation of the K machine.
BVM will no longer pass information to the K monitor.
Fresenius Medical Care 2008K Operator’s Manual
The is a German machine, the series was designed and built in America. Temperature Control Calibration method 3 Completion Of Dialysis Quarterly Hour Preventive Maintenance The key may be used only once.
Periodically check the power cord for damage fraying, over-heating, cuts, scrapes, etc. Page 25 mm Hg and the TMP alarm limits are spread fully open for 30 seconds. Comments dresenius this Manuals Your Name.
Front Panel Controls Blood Recirculation Procedure 2008j audible alarm will sound every two minutes to alert the operator that blood is sensed while the TX Clock is paused. Don’t show me this message again. Sodium Variation System Page Turn machine power off and back on.
Warning A single occurrence is not a problem if the machine automatically resets. After all of the constituents are correctly entered and the name of the concentrate has been chosen, touch the Enter Conc screen button to verify that the concentrate was entered into the list. The Control Panel Chapter 1—Overview The Control Panel The control panel see Figure 3 is located at the top, front of the K machine and contains key pads that allow the user to control the operation of the K machine.
Originally posted by SK8: UF rate will be maintained even with low blood pump speeds.